Yes if downloaded via SquirrelView, but not to MMC. The logger will download until the point that the instruction to download was issued.
Yes, use the advanced function in the download screen.
User specified triggers can activate a total of 4 alarm outputs. As an example; these triggers may be based upon channel conditions.
Triggers can be from analogue, digital channels or time.
No, this can only be achieved via the host
Yes Downloader is supplied on the SquirrelView CD
Thermocouples do not measure absolute temperature, they only measure the temperature difference between the sensing end (the probe tip) and the reference end (where the probe plugs into the logger). To acquire the absolute temperature the logger adds together the temperature difference measured by the thermocouple to the temperature at the reference junction. The reference junction temperature is measured by the Reference Junction channel. To save the user from having to remember to turn on the Reference Junction channel, SquirrelView and the logger do it automatically whenever a thermocouple channel is selected.
The logger is not weatherproof if to be used outside or in damp conditions it will need to be placed in an IP rated enclosure