The foil test involves suspending a strip of foil into various locations around the tank. The foil should not touch the base of the tank and should be held in position for around 1 minute. It should then be removed and there should be an even distribution of perforations and small holes on the surface of the foil.
Another test requires the use of Brownes soil test strips. These are plastic strips which have been contaminated to simulate the contamination which might affect surgical instruments. After running an ultrasonic cycle the strips should be taken from the bath and all contamination should have been removed.
An ultrasonic energy meter can also be used to test the level of ultrasonic activity within the tank.
Please contact Grant for more information on ultrasonic activity testing methods.