If a temperature of below 0C or above 100C is set in the Labwise software, when the setup is sent to the Optima unit, the Optima temperature does not change to that set in the software.
The current work around is to setup a simple 1 segment program using the Programs Editor window.
To set a temperature of 120C (the Optima will need to be configured for High Temp Oil), on the Labwise Programs window.
Set a Target Temp of 120C for segment 1
Set a Duration of 1 minute
Set 'On Program Completion' drop down list to 'Hold at final temperature'
Save the program to the PC (e.g. File name = Set_120C) to send the 'Set_120C' to the Optima, click on 'Bath' and 'Send to Program To Bath' and select the 'Set_120C' file.
The program will start and one minute later the Set temperature will be at 120C.
Note: the liquid may still need time to reach this temperature if it has started of lower than 120C.
At the end of the program the Optima will bleep, to stop the bleeping click on the Green 'Temperature Alarm' status button on the Labwise main page.